Reference adapter plate VAST XXT TL1
The adapter plate forms the central interface between the probe head and the stylus system. The stylus system is fixed via an integrated cube. Zeiss CFX extensions can be used for optimization. The Zeiss adapter plate is read out and recognized on the probe head when it is picked up. For a CNC-controlled automatic stylus system change, the adapter plate (including the stylus system) is placed in the XXT probe socket (620161-8510-000) or the XXT single probe soceket (620161-8511-000), which is located on the storage rack (MSR / ProMax). This adapter plate has a fixed ZEISS Reach CFX1 extension. This reference adapter plate is marked with a red ring. Use both the adaptor plare and the master probe exclusively for calibrating the calibration sphere (reference measurement).
- 4,6 g
Description: ZSH-28-A-REF-TL1-M3 Replaces adapter plate 620161-8158-000 Combinable sensor system: VAST XXT Combinable probe socket: XXT probe socket (620161-8510-000) or XXT single probe socket (620161-8511-000) A maximum weight of 10 g may be suspended from the adapter plate, (adapter plate included). The length of a stylus system may not exceed 125 mm.
XXT TL3 reference adapter plate ZSH-28-A-REF-TL1-M3 with active ID chip
The features of ZEISS measuring machines and the guaranteed machine specifications are only guaranteed when using original ZEISS exchangeable plates. There is a not insignificant negative influence on accuracy and stability if adapter plates from unknown / non-certified manufacturers are used. Check the necessity of performing the DeviceActivator when using a new ZEISS adapter plate. Further information and operating instructions are available at the following link in the Download Center: https://portal.zeiss.com/download-center/softwares/imt